High speed – up to 15,rpm/21, x g * Whisper quiet brushless induction drive * Precise, microprocessor control * Automatic g-force conversion program. Hermle ZM-2 Microcentrifuge from Labnet International, Inc. A high performance microlitre centrifuge, offering a fullrange of features:digital display of time, rpm and RCFtwo acceleration and deceleration.
|Published (Last):||23 January 2012|
|PDF File Size:||2.79 Mb|
|ePub File Size:||19.94 Mb|
|Price:||Free* [*Free Regsitration Required]|
Overcurrent or undervoltage due to power supply fluctuations Action Table of Contents Add to my manuals Add. Enter text from picture: If the green “slow”control lamp is lit, hermld unit accelerates and decelerates very softly below 2 rpm.
Should this happen, the centrifuge, rotor, buckets, adapters and the rotor chamber must be thoroughly cleaned and all broken particles removed immediately. By pressing herm,e button you will print only current page.
hermle z233 m-2 Manuals
Page of 20 Go. The temperature rise depends on the rotor swing-out or angle rotor ,bucket type, ambient temperature, running time and the speed of the rotor.
General Information The Hermle Z M-2 is neither explosion proof nor inert gas shielded and should, therefore, never be operated in explosive-hazardous locations.
Comments to this Manuals Your N-2. Commercially available tubes or bottles made of glass or plastic. The rotor must always be balanced in this fashion prior to operation. Page 14 – Keyboard – Starting the centrifuge – “qu Turn the speed knob 1see figure 4 till hermoe required rotor code appears on the d i s p l a y.
The green control lamps 3 show which mode is activated.
G Hermle Z M-2 Labnet Centrifuge W/rotor | eBay
Preselection Of Operating Time 4. The Hermle Z M-2 is neither explosion proof nor inert gas shielded and should, therefore, never be operated in explosive-hazardous locations. On the digital display “time min-sec” the letters “ro” rotor appears.
Width 28 cm Depth 37 cm Height Operation Operation Operation Operation Operation 4.
Page 19 Error no.: Never stay in the safety zone of 30 cm around the centrifuge or deposit dangerous materials inside this zone during centrifugation.
If this is not done, they could scratch the protective coating of the rotor. There is a distinction between two different kinds of errors. Hetmle have an account?
Refurbished Hermle Z M Centrifuge
Don’t show me this message again. Service And Inspection Of Centrifuge 7. Figure 8 If the green “fast” control m2 is lit, the unit accelerates and decelerates fast. Use only original spare parts! Before operation, secure the rotor lid to the rotor by pressing the snap Before operation, secure the rotor lid to the rotor by pressing the snap Before operation, secure the rotor lid to the rotor by z2333 the snap Before operation, secure the rotor lid to the rotor by pressing the snap Page 20 Error no.: Got it, continue to print.
Your Trusted Source for Used Lab Equipment
To print the manual completely, please, download it. The digital indications on the display will light up when the power is on.
To open the centrifuge lid. Power failure while rotor is in motion Action: Set Or Change Rotor-code -Load the rotor symmetrically by inserting tubes into spaces opposite from each other. During a run and with the centrifuge lid closed you have to press the “preset” key 1hold it and change operating time by turning the knob 3. Open and reclose the lid, restart the centrifuge Error no.: Description – The rotor is not installed securely on the motor shaft.